IntroductionJB8738-1998" vacuum interrupter arc chamber of 3.6-40.5KV AC high voltage switchgear"established by high-voltage switchgear National Standardization Technical Committee, it is prescribed that the internal gas pressure measurement and allowable storage time limit inspection must be executed for any user and producer of interruption arc chamber of high-voltage vacuum switchgear. At the same time, in National Standardization -JB8738-1998, it is also noticed that allowable storage time limit of interruption arc chamber should be 20 years, and internal gas pressure of the vacuum interrupter arc chamber should be less than 6.6×10-2Pa.Inspection methods of Allowable Storage Time of Vacuum Interrupter Arc Chamber: measuring internal gas pressure of vacuum interrupter arc chamber should be with a pulsed magnetic vacuum meter, recode the pressure value P1 (Pa); after put the vacuum meter aside for period of time-t(d), measure the internal gas pressure of vacuum interrupter arc chamber with this vacuum meter again, and recode the pressure value P2 (Pa), holding time t should be less than 7d. By the following formula, the allowable storage time of vacuum interrupter arc chamber can be calculated:VWT-0306 vacuum tester for vacuum switchgear is the latest generation of products of my company, it is the upgrade product in the base of vacuum meter on the foundation of notion from the field user. The vacuum tester has some features-higher measurement accuracy, better stability, more capacity. For vacuum circuit breaker, the method of determining whether or not deterioration of the vacuum tube is inspection with the method of frequency withstand voltage, but this method only distinguish the vacuum interrupter arc chamber of the serious deterioration. When the vacuum of vacuum interrupter arc chamber degrade to 10-2-10-1 Pa, although the breakdown voltage is not reduced, but the interrupter arc chamber has failed. In the vacuum tester for vacuum switch, the new field coils were used, and with the magnetron discharge method the vacuum of the vacuum interrupter arc can be measured without disassembling the interrupter arc chamber. At the same time, the vacuum tester also used computer for synchronous control, data acquisition and processing, and the sensitivity of the vacuum tester in the field test of the vacuum interrupter arc chamber is 10-5Pa. The most prominent feature of this instrument is to use the new excitation coils and data processing methods to achieve the non-demolition measurement of vacuum. The instrument is practical, easy to use, easy to operate, high precision measurement without disassembly, widely used in electric power, steel, petrochemical, textile, coal, railways, and other trades using the vacuum switch.
1.The tester test vacuum bottle without remove it from CB housing.
2.It measure vaccum pressure degree of CB by technic of double inverse magnetron discharge
3.It has good precision, stability and intelligent function.
4.Small size and easy to opreate and carry.Technical Parameters:
1. Test object: All kind of Vacuum Switches.
2. Test Method:Non-disassembly measurement.
3. Use Range: Multi-purpose, suitable for testing multi-style vacuum switches.
4. Test Range: 10-5-10-1 Pa
5. Test Accuracy: 10-5-10-4 Pa, 15%
10-4-10-3 Pa ,15%
10-3-10-2 Pa, 10%
10-2-10-1 Pa ,10%
6. Field Voltage: 1700V
7. Pulse Field High Voltage: 30KV
8. Switch space required: Normal space.
9. Working Enviroment: -20ºC~40ºC
10. Host weight: 12kg
11. Size: 420×320×280(mm)
12. Sampler: Magnetic control coil